The above link is supposed to take you to section 5.4. If it takes you to the top of chapter 5 instead you will have to scroll down to section 5.4 which contains the operating instructions.
Introduction.Intermodulation Distortion (IMD) seems to be the step child of high fi amplifier specifications. I think I know the reason, advertising. Based on what is admittedly a small sample, the IMD figures for a given amplifier are anywhere from 3 to 10 times the THD (total harmonic distortion) figures. So if you were a manufacturer of amplifiers would you post figures that looked much worse than your competitors? I think not. But things don't have to be like that in the hobby world.
I will show in section 1.4 of this work why IMD figures are larger than THD values. Also I will show why IMD is much more objectionable to the ear than harmonic distortion. I think that we as hobbyist should have equipment to measure IMD as well as THD.
Construction of a harmonic distortion analyzer is rather difficult involving some rather high impedance low noise circuits which require good shielding. The level of measurement, typically down to 0.1 % full scale and a frequency range from 5 Hz to 600 kHz in top of the line instruments requires low noise active devices and large value resistors that may not be available off the internet. Automatic null adjustment is a must because manual adjustment is too sensitive to be done day in and day out while keeping one's sanity.
An IM analyzer only needs to be adjusted for 100% and so is much easier to use. In construction some care must be taken to make sure that the signal source does not contain intermodulation products and that the test signal does not leak into the measurement circuits. In the project shown here shielding is accomplished through the use of die cast aluminum boxes to shield the various parts of the instrument from one another. Construction uses prototyping boards so no board etching is needed. No exotic parts are used making the project doable by anyone who has access to the internet.
Chapter 1 Perception and Theory of Distortion.
1.1 The Theory of Measuring THD (Total Harmonic Distortion).
1.1.1 The Input Attenuator and Input Buffer1.2 The Theory of Measuring IMD (InterModulation Distortion).
1.1.2 The Notch Filter.
1.1.3 The Voltmeter.
1.1.4 The Output.
1.1.5 Making the Measurement.
1.1.6 How all this Works Together.
1.1.7 An Alternate Method.
1.2.1 Test Signal Generator.1.3 How We Perceive Distortion.
126.96.36.199 Low Frequency generator.1.2.2 Input Attenuator.
188.8.131.52 High Frequency Generator.
184.108.40.206 Linear Mixer.
220.127.116.11 Output Buffer.
1.2.3 Input Buffer.
1.2.4 High Pass filter.
1.2.5 Detector (Rectifier.
1.2.6 Low Pass Filter.
1.2.7 The Meter.
1.2.8 Instrument Calibration by the DC method.
1.2.9 Calibration by the Two Signal Method.
1.3.1 Harmonic Distortion.1.4 The mathematics of distortion.
1.3.2 Intermodulation Distortion.
1.4.1 Harmonic Distortion and the Gain Polynomial./1.5 Conclusion.
18.104.22.168 The Gain Polynomial.1.4.2 Intermodulation Distortion.
22.214.171.124 The Input Signal.
126.96.36.199 Amplitudes of the Harmonics.
188.8.131.52 The Gain Polynomial.
184.108.40.206 The Input Signal.
220.127.116.11 The Binomial Theorem.
Chapter 2 Theory of Operation.
2.1 Design Considerations.
2.2 Overall Diagram
2.3 Circuit Description.
2.3.1 Control Panel.2.3 Operating Characteristics.
2.3.2 Low Frequency Generator.
2.3.3 High frequency Generator.
2.3.4 Linear Mixer and Output Buffer.
2.3.5 Input Buffer and High Pass Filter.
2.3.6 Linear Rectifier and Low Pass Filter.
2.3.7 Power Supply.
Chapter 3 Parts Acquisition.
3.1 Flat Washers.
3.2 Ordering Parts From DigiKey.
3.2.1 5% Capacitors.3.3 Insulating Washers.
3.2.2 10% Capacitors.
3.4 The Power Transformer.
3.5 Parts List.
3.6 Something I couldn't find.
3.7 Other Parts and Supplies.
Chapter 4 Construction.
4.1 Die Cast Aluminum Box Preparation.Chapter 5 Initial Testing, Calibration, and Operation.
4.2 Circuit Board Preparation.
4.3 Module Construction.
4.3.1 Low Frequency Generator (LF Gen) Board.4.4 Testing, Adjusting, and Troubleshooting, the Modules.
18.104.22.168 IC socket.4.3.2 High Frequency Generator (HF Gen) Board
22.214.171.124 Copper Ground Wire.
126.96.36.199 Grounding Columns.
188.8.131.52 Horizontal Resistors.
184.108.40.206 Vertical Resistors.
220.127.116.11 Semiconductor Devices.
18.104.22.168 External Connecting Wires.
22.214.171.124 Regulator IC Protection Diodes.
126.96.36.199 Drilling Wire Escape Holes in the Box.
188.8.131.52 IC Sockets.4.3.3 Input Buffer and High Pass Filter Board.
184.108.40.206 Copper Grounding Wire.
220.127.116.11 Grounding Columns.
18.104.22.168 Jumpers, Filter Section.
22.214.171.124 Horizontal Resistors, Filter Section.
126.96.36.199 Vertical Resistors, Filter Section.
188.8.131.52 Capacitors, Filter Section.
184.108.40.206 Semiconductor Devices, Filter Section.
220.127.116.11 Jumpers, Oscillator Section.
18.104.22.168 Horizontal Resistors, Oscillator Section. 22.214.171.124 Vertical Resistors, Oscillator Section.
126.96.36.199 Capacitors, Oscillator Section.
188.8.131.52 Semiconductor Devices and Trim Pots, Oscillator Section.
184.108.40.206 Capacitors on Foil Side of Board.
220.127.116.11 External Connecting Wires.
18.104.22.168 Drilling the Adjustment Screw Access Holes.
22.214.171.124 IC Sockets. 126.96.36.199 Grounding Wire.4.3.4 Linear Rectifier and Low Pass Filter.
188.8.131.52 Column Grounding Wires.
184.108.40.206 Jumper Wires.
220.127.116.11 Capacitors on Component Side of Board.
18.104.22.168 Capacitors on Foil Side of Board.
22.214.171.124 Semiconductor Devices.
126.96.36.199 Trimming Potentiometer.
188.8.131.52 External Connecting Wires.
184.108.40.206 IC Sockets.4.3.5 Linear Mixer and Output Buffer.
220.127.116.11 Grounding Wire and Column Grounds.
18.104.22.168 Jumper Wires.
22.214.171.124 A Pair of Diodes.
126.96.36.199 Capacitors on Component Side of Board.
188.8.131.52 Capacitors on Foil Side of Board.
184.108.40.206 Remaining capacitors on component side of board.
220.127.116.11 Semiconductor Devices and Potentiometer.
18.104.22.168 External Connecting Wires.
22.214.171.124 The Leftover Box.
126.96.36.199 Board Preparation.
188.8.131.52 IC Sockets.
184.108.40.206 Grounding Wires and Capacitors on Foil Side of Board.
220.127.116.11 Semiconductor Devices.
18.104.22.168 External Connecting Wires.
4.4.1 Low Frequency Generator.4.5 Chassis Assembly.
22.214.171.124 Setting Up the Test.4.4.2 High Frequency Generator.
126.96.36.199 Voltage and Current Measurements.
188.8.131.52 Filter Frequency Response.
184.108.40.206 Setting Up the Test.4.4.3 Input Buffer and High Pass Filter.
220.127.116.11 Voltage and Current Measurements.
18.104.22.168 Filter Frequency Response.
22.214.171.124 Setting Up the Test.4.4.4 Linear Rectifier and Low Pass Filter.
126.96.36.199 Voltage and Current Measurements.
188.8.131.52 Filter Frequency Response.
184.108.40.206 Setting Up the Test.4.4.5 Linear Mixer and Output Buffer.
220.127.116.11 Voltage and Current Measurements.
18.104.22.168 Filter Frequency Response.
22.214.171.124 Setting Up the Test.
126.96.36.199 Voltage and Current Measurements.
4.5.1 Preparing the Panel/Chassis.4.6 Electrically Bonding the Two Panels.
188.8.131.52 Drilling the Panel.4.5.2 Preparing the Switches..
184.108.40.206 Front Panel Label.
4.5.3 Parts Mounting.
220.127.116.11 Mounting the Switches.4.5.4 Chassis Wiring Part 1.
18.104.22.168 Mounting the Potentiometers.
22.214.171.124 Mounting the BNC Connectors.
126.96.36.199 Power Supply Rectifier and Filter.4.5.5 Chassis Wiring Part 2, Module Installation.
4.5.6 Power Transformer and 120 Volt Primary Wiring.
188.8.131.52a If Both Leads Are Solid Black.4.5.7 Photograph of Finished Product.
184.108.40.206b If One Lead is Solid Black and the Other One is Black With a Colored Stripe.
220.127.116.11 Line Cord Installation.
4.7 The Smoke Test.
5.1 Selecting a Readout Device.Chapter 6 Calibration.
5.2 Initial Tests.
5.2.1 Testing the Signal Source.5.3 Final Assembly.
5.2.2 Testing the Measuring Circuit.
5.2.3 Checking the Input Attenuator.
5.4 Operating Instructions.
5.5 First Measurement.
5.5.1 Discovery of a Design Mistake.
6.1 Calibration to 0.316 volt = 100%.
6.1.1 Adjusting Trimmer in M 4.6.2 Calibration to 1.00 volt = 100%.
6.1.2a Method 1,Two HF Generators.
6.1.2b Method 2, Fancy Schmancy Function Generator.
6.1.2c Method 3, Post Detection DC Level.
6.2.1 Adjusting Trimmer in M 4.
6.2.2a Method 1,Two HF Generators.
6.2.2b Method 2, Fancy Schmancy Function Generator.
6.6.2c Method 3, Post Detection DC Level.
Chapter 7 In Case of Trouble.
7.1 Review of Operating Instructions.Chapter 8 A Distortion Reference.
7.2 Shooting Trouble.
7.2.2 The Power Supply.
7.2.3 The Signal Source.
7.2.4 The Measuring Circuit.
7.2.5 The Input Attenuator.
8.1 The Design Approach.
8.1.1 Circuit Description and Theory Of Operation.8.2 Construction.
8.2.1 Drilling holes in the box.8.3 Initial Test>
8.2.2 Parts Mounting.
8.2.3 Switch Prewiring.
8.2.4 Switch Label.
8.2.5 Final Assembly.
8.4 Measured Values of IMD and THD.
Back to Fun With Tubes.
Back to Fun With Transistors.
Back to Table of Contents.
Back to Top.